|RM C, 13/F, HARVARD COMMERCIAL BUILDING, 105-111 THOMSON ROAD, WAN CHAI, HKfirstname.lastname@example.org|
|Place of Origin:||China|
|Certification:||Certificate of ISO17025|
|Model Number:||ISO 80369-7 Fig C.4|
|Minimum Order Quantity:||1|
|Price:||To be quoted|
|Packaging Details:||safety carton pack or plywood box|
|Delivery Time:||7 working days|
|Supply Ability:||10 pcs per month|
|Brand Name:||Kingpo||Material:||Hardness Steel|
|Warranty:||1 Year||Product Name:||LUER GAUGE|
ISO 80369-7 FIGURE C.4 and C.6 MALE LUER REFERENCE CONNECTOR
ISO 80369-7:2016 cancels and replaces the first edition ISO 594-1:1986 and the second edition of ISO 594-2:1998, clauses, subclauses, tables, figures, and annexes of which have been consolidated and technically revised. Because of this, ISO 594 connections must be tested per ISO 80369-7 requirements. Read: Discussion on the change.
Between ISO 594/2 Fig 7 and ISO 80369-7 Fig C.4
The ISO 594/2 Figure 7 Male Luer Reference Conical Fitting is replaced by ISO 80369-7 Fig C.4 Male Luer Reference Connector. The ISO 80369-7 Figure C.4 is identical to the ISO 594/2 Figure 7 Reference Fitting. ISO 80369-7 Fig C.4 and ISO 594/2 Fig 7 are completely interchangeable. If you request a quote for a new ISO 594/2 Fig 7 Reference Connector from us; we will quote an item which will pass both ISO 594 and ISO 80369-7 requirements. It will be DUO-marked and DUO-certified to both specifications.
Between ISO 594/2 Fig 8 and ISO 80369-7 Fig C.6
The ISO 594/2 Figure 8 Male Luer Reference Conical Fitting is replaced by ISO 80369-7 Fig C.6 Male Luer Reference Connector. The ISO 80369-7 Figure C.6 is identical to the ISO 594/2 Figure 8 Reference Fitting. ISO 80369-7 Fig C.6 and ISO 594/2 Fig86 are completely interchangeable. If you request a quote for a new ISO 594/2 Fig 8 Reference Connector from us; we will quote an item which will pass both ISO 594 and ISO 80369-7 requirements. It will be DUO-marked and DUO-certified to both specifications.
General Construction Details:
Keeping with the requirements of ISO 80369-7; the Reference Connectors we sell are from hardened stainless steel. They have design details which are not specified in ISO 80369-7 and which allow connection to test equipment. These design details vary by manufacturer so be sure you understand the configuration before you order the Reference Connector. The ISO 80369 Reference Connectors are an integral part of Luer needle and syringe testing.
ISO 80369-7 Fig C.4
The ISO 80369-7 Figure C.4 Male Luer Reference Connectors are used for testing Female Luer Connectors for: Leakage; Separation from unscrewing; Stress cracking; and Non-interconnectable characteristics in the ISO 80369-7; Clause 6 specified tests.
Below is the list of tests which are accomplished by using the ISO 80369-7 Fig C.4 Male Luer Reference Connector:
ISO 80369-7 Clause...
6.1: Fluid Leakage
6.2: Sub-Atmospheric Pressure Air Leakage
6.3: Stress Cracking
6.5: Resistance To Separation From Unscrewing