|RM C, 13/F, HARVARD COMMERCIAL BUILDING, 105-111 THOMSON ROAD, WAN CHAI, HKfirstname.lastname@example.org|
|Place of Origin:||China|
|Minimum Order Quantity:||1|
|Price:||To be quoted|
|Packaging Details:||safety carton pack or plywood box|
|Delivery Time:||30 working days|
|Supply Ability:||3 sets per month|
|Electrode Material:||Brass Electrode Head — Platinum||Welding Force:||1.0±0.05N|
|Electrode Distance:||4.0mm ± 0.01mm ，angle: 60°±5||Electrode Pressure:||100~600V（48~60HZ）adjustable|
|Droplet Height:||30~40mm Adjustable||Droplet Size:||44~55 Drops/1cm³ Adjustable|
Tracking tester(Tracking Index Test)is a simulation experiment project using non-flame ignition source in accordance to standard requirements of IEC60112 ,UL 746A ,GB/T4207,GB4706.1,ASTM D 3638-92,etc.
tracking test of leakage current (Tracking Index Test)on the surface of solid insulating material between platinum electrodes in required size ( 2mm × 5mm ),is to exert voltage and drip down a conductive liquid (0.1%NH 4 CL) in required droplet volume at a certain time(30s) and certain height(35mm) ;to evaluate the resistance of electric leakage performance on the surface of solid insulating material in electric field and moist or contaminant medium, under the joint effect;determine Comparative Track Index (CT1) and proof tracking index (PT1) .Dielectric Tester(Tracking Index Tester) is suitable for research,production and quality testing department of lighting equipment,low voltage apparatus,household appliance,machine tool electric appliance,motor-driven machine,electric tool,electronic instrument,electronic gauges and information technology devices;also suitable for insulating material,engineering plastic,other solid combustible material,electrical connectors and accessories industry.
1, electrode material:brass electrode head — Platinum
2, welding force:1.0±0.05N
3, electrode distance:4.0mm ± 0.01mm ,angle: 60°±5
4, electrode pressure:100~600V(48~60HZ)adjustable, voltage decreases to a level less than 8% when short circuit is under the condition of 1.0±0.1A.
5, droplet height:30~40mm adjustable
6, droplet size:44~55drop/1cm3 adjustable
7, dropping time:30s±5s,
8, dropping number:0~9999t adjustable
9, leakage judgement: loop current larger than0.5A and maintain for 2s;if the electric relay cut off the power,it indicates that the specimen is unqualified.
10, test process:automatic control with independent ventilation
11, Reference Standard:IEC60112 ,GB/T4207,GB4706.1
12, studio area:no less than0.5 cubic meter
13, external dimension:1100mmW×700mmD×1300mmH
15, control:single chip Microcomputer touch screen+button type double control system,high accuracy temperature gauge,independent electrical control system,strong anti-interference.
16, operation with high efficiency:One-touch operation ,fast and highly effectively
17, standard integration:program test procedures in accordance with actual test procedures.
* this tester should be placed on a level table
* table size no less than 1500W * 900D * (600-800)H mm
* distance to the wall: > 200mm
* exhaust pipe: diameter of 100mm
* the test room should be independent room equipped with exhaust system and firefighting device
2.1) install the exhaust pipe, tighten the fitting
2.2) check the power supply wiring, ensure it’s correct and grounded
2.3) make up test resolution, and adjust the dropping device accord to standard requirements of:
|Test solution A (CTI A)||395 Ωcm|
|Test solution B (CTI B)||198 Ωcm|
Drops of solution fall on to the specimen at intervals of 30±5s;
The drops shall fall more or less centrally between the electrodes from a height of 35±5mm;
The time for 50 drops to fall on to specimen shall be 24.5±2min;
The mass of a sequence of 50 drops shall lie between 0.997g and 1.147g;
The mass of a sequence of 20 drops shall lie between 0.380g and 0.480g.
Solution prepared should be used within 4 hours to ensure conductivity qualified.
Filling the solution after its full melted, preventing particles clog pipes.
Emptying the liquid when test finished, and wash the pipe with water.
2.4) specimen preparation
The specimen should be flat and clean, flat surface no less than 20mm*20mm, thickness of specimen shall be 3mm or more.
2.5) electrode installation
Adjust the angle(60±5°)and distance(4±0.1 mm) between two electrodes;
Adjust the pressure by move the weight to 1±0.05N；
Fix the screws, and close the window.
Set test parameters
Power on, you can see the welcome interface on touchscreen, it will go operation interface automatically in 3s.
【power】, click power and rotate the voltage regulator, to preset the test voltage
【drip】, click drip, resolution drips, and the counter counts
【exhaust】, click to control the fan
【light】, click to control the light
【short】, click short and rotate the current regulator, preset short-circuit current as 1.0A
【start】, click start, enter testing mode. Tester will stop automatically when the current over 0.5A for 2s