|RM C, 13/F, HARVARD COMMERCIAL BUILDING, 105-111 THOMSON ROAD, WAN CHAI, HKemail@example.com|
|Place of Origin:||China|
|Minimum Order Quantity:||1|
|Price:||To be quoted|
|Packaging Details:||safety carton pack or plywood box|
|Delivery Time:||30 working days|
|Supply Ability:||3 sets per month|
|Glow Wire Temp.:||500~1000℃±2℃ Constantly Adjustable||Thermocouple:||Φ0.5mm Imported Armour|
|Glow Wire:||Φ4 Mm Chromel-filament（Ni80/Cr20）||Broil Into Depth:||7mm±0.5mm|
|Specimen Pressure:||0.95N±0.1N||Test Speed:||18±3mm/s|
Glow wire test is a simulation experiment project using non-flame ignition source in accordance to standard requirements of IEC60695-2-10,UL 746A ,GB/T5169.10,GB4706.1.It makes required material(Ni80/Cr20) and heating wire in required shape(Φ4mm-chromel-filament)heated up to test temperature( 550 ℃ ～ 960 ℃) for 1min and then broil the specimen using required pressure (1.0N) for 30s at vertical direction;to check whether the specimen and bedding material ignite and lasting ignition time to test the ignition fatalness of electrical and electronic finished devices
Test the ignition performance,ignition temperature (GWIT),solid insulating material and other solid combustible material,combustibility and flammability index(GWFI) .
It can be applied to research,production and quality testing department of electrical and electronic devices and component parts like lighting equipment,low voltage apparatus,eletrotechnical instrument,etc.
1, glow wire temp.:500~1000℃±2℃ constantly adjustable
2, broil time:0-999.9s±0.1s adjustable (generally choose 30s)
3, ignition time:0-999.9s±0.1s,automatic record,suspension by wire
4, smother time:0-999.9s±0.1s,automatic record,suspension by wire
5, thermocouple:Φ1.0mm imported armour(type)
6, glow wire:Φ4 mm chromel-filament(Ni80/Cr20)
7, broil into depth:7mm±0.5mm
8, specimen pressure:0.95N±0.1N
9, test speed:18±3mm/s.
10, U head standard:IEC60695-2-10.
11, test procedure:automatic control,independent exhaust system
12, Reference Standard:GB/T5169.10-2006 , IEC60695-2-10:2000,UL746A
13, studio volume:bigger than 0.5 cubic meter,(can be customized as 0.75 or 1cubic meter )
14, external dimension:1100mmW×700mmD×1300mm
15, control:single chip Microcomputer touch screen+button type double control system,high accuracy temperature gauge,independent electrical control system,strong anti-interference.
16, operation with high efficiency:One-touch operation ,fast and highly effectively
17, standard integration:program test procedures in accordance with actual test procedures.
|Glow wire||diameter 4mm±0.04mmNi/Cr(80/20), standard|
|Temp Range||500-1000°C ±2°C continus adjustable|
|Thermocouple||diameter 0.5mm Type K (Ni/Cr~Ni/Al),armored sheath can withstand temperature -40~ 1100°C|
|Searing Time||30s±0.1s (OMRON H5CN-XAN-Z digital timer, 0-9999s, presettable)|
|Light-off Time||0-999.9 s ± 0.1 s, automatic recording, manual timeout|
|Fireouttime||0-999.9 s ± 0.1 s, automatic recording, manual timeout|
|Ironing into depth||7mm ± 0.5 mm|
|Sample pressure||0.95N ± 0.1 N|
|Control||Using single chip microcomputer + touch screen control|
|Power Supply||800VA ,220V, 48-60Hz|
|Consult standard||GB/T5169.10-2006 , IEC60695-2-10:2000,UL746A|
* This tester should be placed on a level table
* table size no less than 1400W * 900D * (600-800)H mm
* Distance to the wall: > 200mm
* Exhaust pipe: diameter of 100mm
* The test room should be independent room equipped with exhaust system and firefighting device
1.1) install the exhaust pipe, tighten the fitting
1.2) checks the power supply wiring, ensures it’s correct and grounded
1.3) connect thermocouple as below:
1.4) connect wire control box as below
Before debugging, please check:
2.1) turn power switch on, the sample carrier come to the left end, left proximity switch light up
2.2) adjust current to minimum, press heating button, and run, sample carrier come to right end, and come back 30s later.
2.3) test sample fixing. Fix the sample tight inside fixture, ensure 7mm spacing in the tolerance. And close the viewing window.
Left: sample carrier small car go left
Right: sample carrier small car go right
Heating: Adjust the current, reach to required temperature. Wait 5min to let the temperature be steady.
Start: Sample carrier automatically moves forward, timers (glow time, Ti, Te) start working, carrier returns when set glow time (30s) reached.
Glow time: 30s
Ti: the duration from the beginning of tip application up to the time at which the test specimen or the wrapping tissue placed below it ignites;
Te: the duration from the beginning of tip application up to the time when flames extinguish during or after the period of application.
Control box: when the sample start burning, press the button on control box, get ignite time Ti; when flame dies, presses the button on control box, get flame extinguish time Te.
When test finished, do not open the viewing window until exhaust for at least 2min.