Payment & Shipping Terms:
|Power:||1KW||Water Supply:||＞10L/min ±5%|
|Dripping Speed:||1mm/min,3mm/min Adjustable||Test Time:||10min, Adjustable|
rain test equipment,
salt spray test chamber
IEC 60529 IPX1 IPX2 Movable Rain Drip Box Ingress Protection Test Equipment
Vertical Drip Rain Testing Machine is designed and manufactured according to IEC 60529 Degrees of protection provided by enclosures (IP Code), this equipment is widely used in lamps, electric and electronic products certification testing of IPX1,IPX2. This device has been widely adopted by domestic and foreign products certification laboratories, the performance of this equipment reaches to the international advanced level.
The program of this test equipment is humanized, users can process IPX1~2 testing without re-setting any parameters manually by choosing the default test options; user also can set the test data to meet different testing standards by choosing the refine test options.
|Power supply||AC220V /50|
|Water supply||The water-flow rates is ＞10L/min ±5%, pure water and without impurities, equipped with water cleaning filter|
|Waterproof test range||IPX1,IPX2|
|Vertical drip rain area||1000mmX600mm|
|Drip needle aperture||Ф0.4MM|
|Distance between holes||20MM|
|Dripping speed (water flow)||1mm/Min,3mm/Min adjustable|
|Drip height||200MM, adjustable|
|Test time||10Min, adjustable|
For the purpose of the tests, the surface area of the enclosure is calculated with a tolerance of 10%.
Adequate safety precautions should be taken when testing the equipment in the energized condition.
Test for second characteristic numeral 1 with the drip box
The test is made with a device which produces a uniform flow of water drops over the whole area of the enclosure.
The turntable on which the enclosure is placed has a rotation speed of 1 r/min and the eccentricity (distance between turntable axis and specimen axis) is approximately 100 mm.
The enclosure under test is placed in its normal operating position under the drip box, the base of which is larger than that of the enclosure. Except for enclosures designed for wall or ceiling mounting, the support for the enclosure under test should be smaller than the base of the enclosure.
An enclosure normally fixed to a wall or ceiling is fixed in its normal position of use to a wooden board having dimensions which are equal to those of that surface of the enclosure which is in contact with the wall or ceiling when the enclosure is mounted as normal use.
The duration of test is 10 min.
Note: When the base of the drip box is smaller than that of the enclosure under test, the latter may be divided into several sections, the area of each section being large enough to be covered by the dripping water. The test is continued until the whole area of the enclosure has sprinkled for the specified time.
Contact Person: Ms.Lynette Wong