Payment & Shipping Terms:
|Standard:||IEC 60884-1 Fig 18 And Fig 19||Purpose:||Verification Of Maximum Withdrawal Force|
electrical socket tester,
power socket tester
IEC 60884-1 Fig 18 and Fig 19 Verification Force Plug Socket Tester Apparatus
This Tester is designed to meet the exacting requirements of major international standards such as IEC 60884-1 Fig 18 and Fig 19. It is used to test whether the maximum force of pulling the plug from the socket and the minimum force of pulling the single plug pin from the plug bush is in the range of the standard required. The clamping system is easy to operate.
The top plate swivels up from a horizontal position to a vertical orientation and the clamping device can be rotated in the top plate. These features allow the device to be set up in many different configurations to meet different standard requirements. Standard plugs and weights are supplied to perform testing on sockets rated up to 16A.
Pin gauges and the appropriate weights for testing minimum withdrawal force are sold separately. The required dimension of the pin gauge needs to be specified by the customer.
Test plugs made from hardened steel with roughness not exceeding 0.8mm.
Top plate pivots from horizontal to vertical.
1, It is equipped with 10A ambipolar plug ,the three pole plug of 10A and 16A and the testing standard plug pin of 10A,16A.
2, It is also equipped with a set of weight and a set of clamping system.
|Ratings for the accessory||Number of the poles of the accessory||Withdrawal force N|
|Multi-pin gauge maximum||Single-pin gauge minimum||Single-pin gauge maximum a|
|Up to and including 10 A||2||40||1.5||17|
|Above 10 A up to and including 16 A||2||50||2.0||18|
|More than 3||70|
|Above 16 A up to and including 32 A||2||80||3.0||27|
|More than 3||100|
|a These withdrawal forces are only for testing the resilient earthing contact assembly of a plug.|
Contact Person: Ms.Lynette Wong