Turntable -Ingress Protection Test Equipment , IEC Test Equipment
The rotating table is manufactured in accordance with the standard of Clause IEC60529 testing purpose.The loading platform can be inclined 15°to meet with IPX2 inclined testing requirement.
1. Turntable installation
Loosen the screw on the upper part of the supporting column, rotate the upper support stage, can adjust the height of turntable.
The bottom is provided with four supporting feet, and the supporting feet is adjusted to make the turntable perpendicular to the ground. Two screws were installed opposite at the bottom, loosen the two screws, can tilt the turntable. Tightening the screws according to the screw hole position after tilt, the support stage with an angle of 15 degrees to vertical turntable.
Place the sample to be tested on the turntable, adjust the tilting screw at the bottom of the turntable as needed, and tilt the turntable 15°. Press the start button and adjust the speed regulator according to the desired speed. If the sample is too heavy, the upper support stage can be removed and placed directly on the underside aluminum plate.
||continuous rotation, Every 150 s turn 90 °,Every 300 s turn 90 °
||≤150kg(Underside aluminum plate)
||PLC + touch screen