Payment & Shipping Terms:
|Standard:||IEC 61347-1:2012, Clause 14.6||Working Voltage:||<1000Vac|
|Working Frequency:||50/60Hz||Working Current:||<500Aac＜500Aac|
|Flow:||160A +10% -0% Current Constantly||Function:||Current Test Device|
temperature rise test equipment,
led lamp testing equipment
IEC 61347-1-2012-Figure 3 Test Circuit for Controlgear
This tester is a current test device developed according to the IEC 61347-1:2012 standard, Article 14.6 requires current compliance; through the tester current measurement, it can determine whether the test conditions meet the "standard".
1.1. According to the IEC 61347-1:2012 standard, clause 14.6 a requires that the circuit be connected, as shown in figure (1) :
1.2 Adjust the oscilloscope input display ratio to X1, Y-axis adjustment to 1V/div, X-axis adjustment to 20mS/div, and the synchronization status is set to normal synchronization.
1.3 For example: connect the power supply, the oscilloscope waveform is the current of the test device, and the waveform is shown in figure 2.
Current reading method: peak voltage: 1.9V, peak current 190A (current to voltage ratio 100:1) RMS voltage: 1.34V, RMS current is 134A
According to the requirements of IEC 61347-1:2012, power supply and test equipment can be qualified to flow 160A +10% -0% current constantly.
2.1 According to IEC 61347-1:2012 standard, Clause 14.6-b connects the tested products, as shown in Figure below
3.1 According to IEC 61347-1:2012 standard, Clause 14.6-C, connect to the power supply for testing, oscilloscope setting method and current reading method is the same as 1.2,1.3.
1, working voltage: <1000Vac
2, working frequency: 50/60Hz
3, working current: <500Aac＜500Aac
Contact Person: Ms.Lynette Wong