High Accuracy Plug Flexing Tester With Touch Screen / Alarm
|Place of Origin:||Dongguan,China|
|Certification:||international recognized test lab, with CNAS and ilac.MRA marks.|
Payment & Shipping Terms:
|Minimum Order Quantity:||1|
|Price:||need to be quoted|
|Packaging Details:||safety carton pack or plywood box|
|Delivery Time:||25 working days|
|Supply Ability:||3 sets per month|
|Output Power:||1000W||Cores Of The Specimen:||2—5 Cores|
|Working Channel:||1||Rotating Speed:||5-60 Cycle/min(can Be Adjusted)|
High Frequency Spark Tester,
Bend Test Machine
Chapter one: introduction of the instrument
1.1 Characters and main use
This tester is used to perform the fatigue test on the AC and DC power cord and power cords lead. This is the necessary equipment for the QC department and Lab of the related industry.
The rotating speed and rotating angle can be preset freely, and it will alarm once the specimen was damaged, it has many good features, such as: high accuracy, stable, and high efficiency, which is a perfect tester for fatigue test.
1.2 Design standards
UL817, IEC60884, VDE0620, GB2099 and so on.
1.3 Technical Parameter
|4||Cores of the specimen||2—5 Cores|
|6||Flexing angle range||50-9999°can be adjusted)|
|7||Flexing precision||+/- 1 °|
|8||Rotating speed||5-60 cycle/min(can be adjusted)|
|9||Test times range||0-999999|
|10||Counter precision||+/- 1 times|
|11||Weight||1 kgs/2 kgs(can be equipped according to requirement)|
|12||Failure alarm||The machine will alarm and the touch screen will have indication once the specimen is damaged.|
|13||Qualified indication||The machine will alarm and the touch screen will have indication once the real testing times meet the preset test times.|
|15||Net weight||Approximately 10 Kgs|
Chapter two: installation introduction
2.1 Mounting and dismounting noticeable points
2.1.1 Check the instrument if damaged in transit;
2.1.2 Check the instrument if invert；
2.1.3 Check if the specification on the nameplate of the instrument accord with your order;
2.1.4 Check if the packing list accord with real object.
If above abnormal happened, please don’t use this instrument, and timely to contact with agency or our company.
2.2 Safety rule
2.2.1 Before using this machine, please read manual in depth, learn related safety signs and operation steps, to ensure safety.
2.2.2 Check if the real power supply accord with it in nameplate and make sure the power well grounded.
2.2.3 Layperson forbidden to open the door or rear cover of the machine, to avoid electric shocking.
2.2.4 Instrument screen cannot be scraped by sharp objects.
2.2.5 User cannot modify the circuits or elements, otherwise the guarantee will be invalided, caused any consequences will not be afforded.
2.3 Installation requirement
2.3.1 The instrument should be placed on horizontal and fixed work stations.
2.3.2 the instrument used ambient temperature should be 25±10℃, related humidity 50±30%RH
2.3.3 The instrument should be far away from heat source and corrosion source.
2.3.4 The instrument should be have well grounded power supply AC 220V/50Hz±10%.