IEC 60695-10-2 2014 Ball Pressure Tester With Test Specimen Support Hot Deformation Capacity
|Place of Origin:||China|
Payment & Shipping Terms:
|Minimum Order Quantity:||1|
|Price:||To be Quoted|
|Packaging Details:||satety plastic tool box or plywood package.|
|Supply Ability:||25 per month|
|Standard:||IEC 60695-10-2:2014||Ball Diameter:||5mm (R2.5mm)|
|Application:||Heat Testing In Household And Similar Plug Sockets||Test Pressure:||20N ± 0.2N|
|Material:||Stainless Steel||Test Stand:||100mm (height) X 50mm (diameter)|
articulated finger probe,
ul finger probe
IEC 60695-10-2:2014 Ball Pressure Tester With Test Specimen Support / Hot Deformation Capacity Tester
This Ball Pressure Tester applicable to assessment of heat-resistant fro all insulation materials of electrical enclosure, external components and plastic. Trial is that the sample placed on the platform of inside constant temperature under 125±2Degree, coupled with the ball-side of tester suppressing the specimen, remove the specimen and put into the cold water for 10 seconds after 1 hour, and then measuring Sag values, indentation of ball diameter shall not exceed 2mm.
Adopts steel materials, nickel plating has excellent heat-resistant
Responding to handle the bow with installation and threaded holes
Responding to the pressure precision ball (according to 5mm ISO3290)
Adopts thermocouple test
electrical and electronic equipment, components and partsCeramics and other solid insulation materials
Meet the IEC60695-10-2(2003), IEC 60884-1, IEC60320-1, IEC60335-1, IEC60598-1, IEC60950-1.
GB 2099.1, GB4706.1
Pressure: 20N (including loading and unloading equipment and ball)
Ball Pressure Diameter: R = 2.5mm
Test Stand: 100mm (high) x 50mm (diameter)
The test specimen support shall be such that
1. it rigidly supports the test specimen in a horizontal position.
2. It has sufficientstrength to support the loading device.
3. It has a smooth flat surface.
4. It has sufficiently large mass to prevent a significant reduction in temperature of the apparatus during the installation and removal of the test specimen from the heating oven.
NOTE It has been found useful to mount a separate thermocouple in the certer of the test specimen support approximately 3mm below the surface to check that the temperature of the test specimen support does not significantly deviate from the test temperature.
8.2 Heating oven and test apparatus setup
Conduct the test method in air,in a heating oven(see 5.3) at a temperature specified in the relavant specification(within a tolerance of ±2℃) measured within 50mm of the approximate center of the test specimen.The test specimen support and loading device shall be preheated at the test temperature for a minimum of 3h